In a past post, I’ve attached a picture of the load I was using for speaker amplifier testing. I have a box full of these 50W wire-wound resistors and a heatsink (HS) I’ve tapped to be able to attach these resistors easily. I was simply connecting as needed for the specific case. In practice, I rarely change the default 8×2-ohm resistors which are split into 2 loads of 8-ohm each. When I needed to dissipate significant power I would normally point a fan at that HS and be done with it. However, this wasn’t very convenient, and I wanted something more “user friendly” to replace it, this is what will be described in this post.
Over the past year and a half since posting the series of pages about the measurement pre-amplifier I’ve designed and built, I’ve received emails from multiple people who were interested in building the pre-amp. I’ve happily shared with them the remaining boards I’ve had from that first batch I ordered at the time. These few boards were all given out, and I’ve therefore ordered a few extra boards to be able to keep offering these boards to people who would like to build such an instrument for themselves. Unlike the first batch, this time I’ve printed boards that fixed the issues I’ve reported when building my own unit and were fixed by a “bodge” over the original board. These modifications to the board design (and the writing of this post) were all done well over a year a go when I’ve originally assembled my unit, but I didn’t see a need to post them until this point in time.
This post is meant to share the updated schematic, as well as to offer additional information that can be of help to people who would like to assemble such an instrument. I didn’t make any functional changes to the pre-amp at this revision, therefore I will offer no additional measurements in this post.
Over the past few months a few different Hafler amplifiers passed through my bench. First it was a combo of a 915 pre-amplifier with a 9303 power-amplifier to accompany it, and some time later I got a 9505 power-amplifier. If you’ll have a look at the schematic you will see that while the pre-amp isn’t too exciting, the power amplifier has a few interesting points in its schematic. So I’ve decided these amplifiers are interesting enough to warrant a short post about them with some pictures and measurement results.
In this post I’d like to briefly share my thoughts on a subwoofer amplifier board I’ve purchased from AliExpress a few weeks ago. I have purchased it as a cheap solution to test an in-wall subwoofer I’m building (2x10inch woofers model R1S4-10 from RF in sealed enclosures). I looked for an amplifier that will have ~400W output power into an 8ohm load (bridge connection), although I doubt I will ever need this much power, or even half that. I’m actually more interested to see how much output power I can get before distortion gets out of hand, with the hope I could get ~200W before onset of clipping. I would obviously much prefer to have a plate amplifier for this, but I haven’t found one that seemed to meet all my needs for a reasonable cost, so I went with plan B. As all temporary things, if it works well it will probably be turned into a fixed solution, so I was interested in seeing how it will perform on the test bench.
This is part 4 in the series of posts discussing the (audio) measurement pre-amplifier project. In part 1 I’ve covered the motivation for this project along with the circuit schematic and detailed circuit description. In part 2, I have gone through the board layout consideration and showed the assembled boards. In part 3, I have gone through measurement results of the assembled pre-amplifier board, as well as some circuit modifications to extend its performance. In this post, part 4, I will briefly show the assembled unit, along with slight discussion of external and power supply coupling into the signal.
As with many of my recent projects, I stuck to PCB’s for the front an rear panels of the pre-amplifier. The benefits are clear, its cheap, its very easy to design in the same software tools used for all of my circuit designs, and it offers electrical shielding due to the internal copper layers that are available to us. Unlike in my previous builds, this one is significantly larger, has very large holes, and even square cut-outs. Therefore, I wasn’t sure how well it will come out. To minimize the chance of an error I’ve printed the panels on a piece of paper and measured it in place before placing the orders. You don’t want to spend a few 10’s of $’s, and wait for a few weeks before you realize you’ve made a mistake 🙂 Continue reading “Audio Measurement Pre-Amplifier – Part 4 – Casing the Pre-Amplifier”
This is part 3 in the series of posts discussing the (audio) measurement pre-amplifier project. In part 1 I’ve covered the motivation for this project along with the circuit schematic and detailed circuit description. In part 2, I have gone through the board layout consideration and showed the assembled boards. In this post, part 3, I will show some of the measurement results of the assembled boards. I will start with describing what it is I would like to measure, and how I plan on measuring it, including the limitations of the measurements I can make with the gear available to me. Then I will show the relevant result and discuss them.
The measurements I plan on performing can be split into 3 different groups. The first has to do with linearity of the pre-amplifier, to measure how much distortion it will have. Next are the noise measurements, as I want to verify the input referred voltage noise of the pre-amplifier to make sure it meets my target figures to allow measurement of low noise voltage regulators (and other devices). Finally are the “other” tests such as the accuracy of the True-RMS reading, the voltage limits of the output protection circuit, and so on.
This is part 2 in the series of posts describing the audio measurement pre-amplifier project. In part 1 I’ve covered the motivation for this project along with the circuit schematic and detailed circuit description. In this post, part 2, I will discuss the next steps related to the board design and assembly. This part won’t be as long and the first (I hope), but I would like to share some of the consideration I’ve made when laying out the board design.
The first step was deciding on a case size and layout for the front panel, as this will set some constraints on board dimensions and placement of connectors/switches/LED’s. I wanted to use a case that will be made of aluminium to use it as a shield, as at the highest gain setting the pre-amp has 60dB (X1000) of gain which makes it very sensitive to coupling from external signals. I also plan on placing the completed pre-amplifier on my work bench, so I wanted something that is relatively compact, but isn’t too cramped so that it isn’t comfortable to use. Something similar (or slightly smaller) than a bench DMM seemed like a good size for this as I would be able to stack it on top of my other instruments. The plan was to have all the relevant connectors and switches at the front, along with some LED’s for visual representation of the selected range, and a panel mounted voltmeter. Placing it all in a single row seemed impossible, or at least very uncomfortable to use. Therefore I’ve decide to split this into 2 different rows (heights). This put a constraint on the minimum height of the case, and meant I will have to split the design into 2 boards to support this since I don’t want to solder any wires. The schematics posted in part 1 of this series already represented this split board solution, with the second board used mostly for range selection.
As some my other posts show, I have been spending a significant amount of my spare time over the past few months on audio measurements related stuff. This included a low distortion oscillator, a notch filter to go with it, as well as modifying the EMU 0404 USB to extend its performance. One other item that has been in the works for quite a long time, is an audio measurement pre-amplifier. The motivation for this work is quite straight forward, I needed to find some way of turning the sound-card I’m using into a versatile measurement tool to do general audio measurements. The most significant limitation with sound-cards is their limited input voltage range, as most audio amplifiers put out voltages that are significantly higher than what you can safely feed into a sound-card. Indeed, this is what most people would use such instruments for. However, this is actually just a portion of what such a pre-amplifier could be used for.
This post will the first part of a series of posts that will describe my take on a measurement pre-amplifier. I will describe the motivation (requirements), the circuit design and implementation, measurement results, and more. I will try to make this as informative as I can, and share some of the reasoning behind design decision. I think this can be of value for both people who would like to understand the circuit better, and people who would like to modify the circuit to better suit their needs.
One important tool that can help extend the capabilities of a distortion measurement setup is a notch filter. The logic behind it is fairly simple, if we are only interested in the distortion components, why should we even feed the fundamental frequency into the measurement setup? By eliminating it (or simply attenuating it sufficiently), we can reduce the harmonic distortion generated by the test equipment as a result of the large tone, effectively extending its capabilities for harmonic distortion measurement. There is obviously more than one way of doing it, and in this post I will only describe one way which was a good match for my needs.
I wanted to create a small box that would implement this function for my needs to allow me to extend further the THD measurement setup I have. In its simplest form, using the EMU 0404USB I’m able to measure THD of ~0.001% at 1KHz. By using an external low distortion 1KHz oscillator I was able to extend this down to ~0.0004%. However, I was looking for a way to get down to 0.0001% to allow measurement of high quality DAC’s. Since I know the external oscillator I use has sufficiently low distortion to support these figures, I needed a way to reduce the distortion caused by the input stage and ADC of the EMU. I have considered trying to hack the EMU and improve its input stage, but I expect the ADC will limit me before I can reach the target performance. Therefore I went with the option of removing the fundamental frequency from the signal before feeding it into the EMU, to reduce the distortion it generates.
One tool I use quite often when testing amplifier is a dummy load. Because of that I have a large box of high power (50W) resistors, and a large heat-sink that is tapped for easy attachment of the resistors to it. I typically have 8 resistors of 2ohm each, connected according to the requirement of the measurement i’m doing at the moment. More often than not, they are wired as 2 independent 8ohm resistors to measure speaker amplifiers (see Fig. 1). However, when I need to measure headphone amplifiers, I typically only need lower power loads, and therefore use a couple of resistors from the spare parts box. This got frustrating over time, soldering the resistors to a TRS plug, then soldering/clipping on a couple of wires to the scope/other test instrument. Therefore I’ve decided to do a small side-project of building a simple dummy load box for headphone amplifiers testing.