Over the past few months a few different Hafler amplifiers passed through my bench. First it was a combo of a 915 pre-amplifier with a 9303 power-amplifier to accompany it, and some time later I got a 9505 power-amplifier. If you’ll have a look at the schematic you will see that while the pre-amp isn’t too exciting, the power amplifier has a few interesting points in its schematic. So I’ve decided these amplifiers are interesting enough to warrant a short post about them with some pictures and measurement results.
A few months ago I came across a faulty programmable power-supply (PS) with a 60V/12A maximum rating on each of its two channels. The exact model is DTPS6012 from Horizon, a company I’m familiar with as I’ve used and owned a few of their linear PS’s (such as the DHR40-1). The problem that was observed during initial check at the seller’s location was that upon power up one of the channels behaved as expected, while the other wasn’t regulating the output voltage. The voltage just kept on rising until it was ~10% over the 60V rating, at which point the over-voltage-protection (OVP) kicked in and switched off the entire unit except for the front panel. Because the unit was faulty the price was quite low, so I’ve decided to purchase it and try and fix it. At the very least this could be an opportunity to have a look inside and learn how these things were built back then.
I should note that such a high power rating PS is more than I will probably ever need for my projects. However, I have had some projects in the past where the 2x3A rating of my existing PS’s wasn’t enough, even when I’ve used two such units. Therefore, a more capable PS, even if its not as low noise and ripple, is always welcome. Additionally, as I’ve noted earlier, I have owned and used elsewhere other PS’s from Horizon. I was always happy with the build quality and performance, especially for the price these things could be had on the used market.